Re: conflicting results
Jun 14, 2003
Greetings, A follow up on my earlier posting. I had a redraw done on my viral load, and, good news, it is returning back in the right direction. From 1,200 to 448. (being undetectable for 3 years). 1 week time between the 2 draws. I found out that the lab, has had "glitches" causing the jump in viral load for many. Are technical problems in reading the assays common? All my other numbers were very conflicting with the "glitched" viral load draw. Percentage, and Tcell all headed in the right direction. I just wanted to post these results, and say that "technical" problems do happen even in the best clinics. My Dr., says I'm fine and not to worry, especially since I've been 100 compliant with the Sustiva and Combivir. Needless, until I found out about the "glitches" I've had many sleepless nights this past week!!
Response from Dr. Holodniy
Yes, glitches happen. Sometimes it's the technician, sometimes it's a new lot number of reagents. That's why it's important to always retest when VL results look funny.
Get Email Notifications When This Forum Updates or Subscribe With RSS
- What Is The Risk Of Getting HIV From Cut On Hand?
- Is Sore Throat A Symptom Of Ars?
- Burning Penis After Condom Broke Worried I Have HIV
- Itchy Scrotum After Touching Vaginal Fluid Sign Of HIV AIDS
- Sore On Penis After Oral Sex Hooker What Are The Chances Of HIV
- Thrush After Happy Ending Worried I Have HIV
This forum is designed for educational purposes only, and experts are not rendering medical, mental health, legal or other professional advice or services. If you have or suspect you may have a medical, mental health, legal or other problem that requires advice, consult your own caregiver, attorney or other qualified professional.
Experts appearing on this page are independent and are solely responsible for editing and fact-checking their material. Neither TheBody.com nor any advertiser is the publisher or speaker of posted visitors' questions or the experts' material.